Powerful, All-in-One Solution
With the NanoBrook Omni, measure samples in nearly every possible suspension environment from high salts such as PBS and sea water to organic solvents and aqueous solutions. Particle and protein sizing, with the highest sensitivity, utilizing three measurement angles for truly unbiased measurement results. The NanoBrook Omni utilizes a backscattering angle as well as 90° and 15°.
The NanoBrook Omni particle size and zeta potential analyzer incorporates all you need for fast, routine, sub-micron measurements of size and zeta potential. Based on the principles of Dynamic Light Scattering (DLS) for particle sizing and distribution, and based on doppler velocimetry (electrophoretic light scattering or ELS) for zeta potential measurement, most measurements only take a minute or two. The instrument also includes Phase Analysis Light Scattering (PALS) measurements for samples with low mobilities.
Features at a glance
- Rapid and accurate protein & nanoparticle size distributions
- Multimodal & unimodal size distribution software
- ISO 13321 and ISO 22412 compliant results
- Range: < 0.3 nm to 10 µm
- Three measurement angles: 15°, 90°, and 173°
- Ideal for fast, routine sizing applications in research or quality control
- High power 40 mW diode laser
- Dynamic light scattering at 173° and 90°
- Temperature control, -5 °C to 110 °C
- Compact bench top unit, USB connection
- Molecular weight determination (relative and absolute through Debye plot)
- Zeta potential for the difficult cases
- For proteins, peptides, mAb, RNA and other biological samples
- For zeta potential in organic solvents
- For oily or viscous media
- For high-salt suspensions
- For samples near the I.E.P.
- 1,000 times more sensitive than other techniques
- Disposable cuvettes, no contamination or alignment
- Built in automatic procedures and parameters (SOP)
- Easy to use
Simple fluids like water (low viscosity), glycerin (high viscosity) are Newtonian and exhibit viscosity effects, the dissipation of energy when particles move in such fluids. But dissolve macromolecules in these liquids –synthetic or biopolymers—and networks can form. In addition to viscosity effects, there are now elasticity effects, the storage of energy when embedded particle move. By following the mean square displacement (MSD) of tracer (probe) particles in such fluid and microrheological properties such as η*, the complex viscosity, G″, the viscous loss modulus, and G′, the elastic storage modulus, can be determined as a function of frequency.
Measurement of the autocorrelation function (ACF) using DLS techniques yields the MSD of tracer particles, which, under the right conditions, can be used to determine η*, G″, and G′ over a range of frequencies much higher than mechanical rheometers can attain. Much smaller sample volumes, in the microliters, are possible compared to mechanical instruments. Finally, since strains result from the thermally driven motion of tracer particles, these much smaller strains allow the study of fragile samples. The study of viscoelasticity in aggregating dilute protein solutions is a prime example of the benefits of DLS microrheology.
Surface Zeta Potential – Principles of Operation
The Surface Zeta Potential feature allows the user to measure the electrical charge on materials like coated glass, plastic, tape, or other flexible surfaces. A series of measurements are taken on probe particles at different distances from a surface and the Surface Zeta Potential is calculated as shown:
The NanoBrook Omni is the one true “All-in-One” solution to particle and protein characterization.
Rapid, Reliable, and Accurate Analysis
The new NanoBrook Omni particle size and zeta potential analyzer incorporates all you need for fast, routine, sub-micron measurements of size and zeta potential. Based on the principles of Dynamic Light Scattering (DLS) for particle sizing and distribution, and based on doppler velocimetry (electrophoretic light scattering, ELS) for zeta potential measurement, most measurements only take a minute or two. The instrument also includes Phase Analysis Light Scattering (PALS) measurements for samples with low mobilities.
Three Scattering Angles:
Measurements of traditional colloids are usually made at 90° scattering angle due to the unbiased results measured. For nanoparticles and proteins, IgG and peptides, these < 50 nm samples can be measured using the backscattering angle (173°) for best S/N and reproducibility of measurements. Finally the 15° detection angle can be selected for added sensitivity with aggregation measurements. Zeta potential measurements are always performed using the 15° detection angle to minimize diffusion broadening.
Principles of Operation – Sizing
Dilute suspensions, on the order of 0.0001 to 1.0% v/v are prepared, using suitable wetting and/or dispersing agents, if required. A small ultrasonicator is sometimes useful in breaking up loosely-held agglomerates. At 173° sample volume may be reduced to 50 µL with a polystyrene, U-shaped, disposable cuvette and the sample is recoverable. At 90° square polystyrene or glass cells (two or three mL) are used, one as small as 10 µL (non-disposable). In addition, disposable, glass round cells with reusable Teflon stoppers are used for aggressive solvent suspensions. In all case, just a few minutes are required for the sample and cell to equilibrate with the actively controlled temperature environment inside the NanoBrook Omni.
Perhaps the most important benefit for users is the reproducibility from sample-to-sample, operator-to-operator, and instrument-to-instrument. The table below shows the results obtained for a polystyrene latex. Notice the Std. Error of the Effective (or Average) Diameter is less than 1% of the mean of the 10 repeated runs; the duration of each run was only five minutes. Such excellent repeatability is typical of the 90Plus for even routine QC analysis and ensures product quality time-after-time.
|Test Results from Brookhaven Instruments Particle Size Analyzers(diameters, in nm)
The following table shows several aspects of the inter-instrument repeatability found with routine Brookhaven dynamic light scattering equipment. To test the reliability of the information obtained with Brookhaven Instruments a customer subjected five instruments in five locations with different operatorsand using four different samples to examination. The tabulation below shows excellent reproducibility with recognized standard materials and superb internal agreement. You can be confident of the results you get with your NanoBrook Omni!
The NanoBrook Omni particle size and zeta potential analyzer offers three choices. For routine determinations an average diameter (Eff. Dia.) and a measure of the distribution width (Polydispersity) are sufficient for many applications. This is illustrated above in Figure1 for the latex with a narrow size distribution. The second choice is to fit these values to a lognormal distribution, allowing the user to visualize the size distribution and to interpolate cumulative and differential results at 5% intervals.
Figure 2: Results from Test Bimodal Sample on NanoBrook Omni (diameters, in nm)
Figure 2 above shows an example of the third choice, suitable for more complicated, multimodal size distributions. Here, a numerical algorithm, including Mie theory, is used. These results are for a mixture of known latex particles. Positions of the measured particle sizes on the accompanying graph are in excellent agreement with the known sizes of 92 and 269 nm.
During a measurement, the display can be switched interactively between any one of these — correlation function, lognormal, or multimodal — each shown “live” as data are accumulated. The live display is particularly useful in determining the end-point of a measurement where multimodal distribution shape may be important
NanoBrook Omni – Phase Analysis Light Scattering
For measurements of very low mobilities, the Brookhaven NanoBrook Omni is the answer; the only answer! With concepts developed at Bristol University and Brookhaven Instruments, the NanoBrook Omni determines zeta potential using phase analysis light scattering: A technique that is up to 1,000 times more sensitive than traditional light scattering methods based on the shifted frequency spectrum.
Electrostatic repulsion of colloidal particles is often the key to understanding the stability of any dispersion. A simple, easy measurement of the electrophoretic mobility “even in nonpolar liquids” yields valuable information. Measurements made in water and other polar liquids are easy and fast with the NanoBrook Omni. Such measurements cover the range of typically ± (6 to 100) mV, corresponding to mobilities of ± 0.5-8×10-8 m2 /V·s. The NanoBrook Omni covers this full range, of course, and extends it by a factor of 1000 in sensitivity!
Principles of Operation – Zeta Potential
The NanoBrook Omni utilizes phase analysis light scattering to determine the electrophoretic mobility of charged, colloidal suspensions. Unlike its cousin, Laser Doppler Velocimetry (LDV, [sometimes called Laser Doppler Electrophoresis, LDE]), the PALS technique does not require the application of large fields which may result in thermal problems or denaturation. This is due to the fact that the measurement analyzes the phase shift. The particles need only to move a fraction of their own diameter to yield good results. In salt concentrations up to 2 molar and with electric fields as small as 1 or 2 V/cm enough movement is induced to get excellent results. In addition, the Autotracking feature compensates for thermal drift.
Simple and clear presentation
Figure 3 above shows the results of an actual experiment with a NanoBrook Omni instrument. The important parameters and results are seen at a glance. The excellent agreement of the five runs in this experiment is obvious as is the match of experimental curve (red, bold) and it’s fitted version (red, thin). As with all Brookhaven instruments the user can simply produce a customized report.
The software can be easily customized to display the columns needed for a quick review of the important parameters as shown below.
Comprehensive Information – ELS
The NanoBrook Omni measures complete electrophoretic mobility distributions in seconds including multimodals. An example of bimodal zeta potential sample can be seen on the result screen from analyzing a created mixture of charged particles.
In Figure 4 below the results of analyzing a mixture of alpha and gamma Aluminas in 1 mMolar KCI at pH10 is displayed. The left peak is identified with the green cursor and shown to have a zeta potential of -20.54 mV. If the other peak is chosen the value given is -5.00 mV. The ability of the NanoBrook NanoBrook Omni to provide this information distinguishes it from other methods which provide only an ensemble average.
Something more challenging – PALS
Of course the NanoBrook Omni can quickly and easily yield results from all “regular” samples but its real strength is in the difficult cases and to demonstrate the performance of this premium instrument where others fail, we offer the follow table.
Multiple Sample Types
Table 1 below shows a variety of difficult to measure samples, all of which were easily measured with the NanoBrook ZetaPALS. Some were measured in high salt concentration; some in low dielectric constant non-polar solvents; and one in a viscous liquid.
|Electrophoretic Mobilities Determined with the NanoBrook Omni
(units 10-8 m2 /V·s)
||2.51 ± 0.11
||2.53 ± 0.12
||Electrophoretic mobility standard.
||-1.081 ± 0.015
||-1.08 ± 0.02
||Dispersed in physiological saline
||0.013 ± 0.0015
||Dispersed in dodecane
||0.255 ± 0.010
||Dispersed in toluene – not dried
||0.155 ± 0.011
||Dispersed in toluene – dried
||-0.503 ± 0.0015
||Dispersed in ethanol
||-0.025 ± 0.002
||Dispersed in PEG – viscous
||-0.73 ± 0.04
||Dispersed in 2.0 M KCl – High salt
Biological samples such as proteins, antibodies, peptides, DNA/RNA are easily denatured by electrical fields. Brookhaven’s NanoBrook Omni can successfully measure the mobility of biological samples with typical voltages from 2 to 4 Volts. In Figure 3 above, Lysozyme was measured with 2.5 volts applied.
Aggressive solvents such as DMF, THF, DMSO, MEK, etc., are easily accommodated by the Brookhaven NanoBrook Omni system with the use of our special solvent resistant electrodes and glass sample cells. The extension of zeta potential measurements in the realm of such systems is just another standout property of the Brookhaven NanoBrook Omni.
Usual solvent? If your solvent is unusual then its dielectric constant is probably unknown. In this case our BI-870 Dielectric Constant Meter will quickly and accurately provide the information necessary for a zeta potential measurement.